How to resolve the SATURATION Problem on Rosemount Transmitters
By W.L. Mostia on 13 August, 2018 - 12:03 pm
I would try the following to see if you can get any additional clues to help you out: 1. The...
By Amar DAMENE on 13 August, 2018 - 2:58 am
Dear all, Yes, I agree with you. I have read what the 3051 manual says about saturation level...
By Amar DAMENE on 12 August, 2018 - 3:06 am
Dear William, I will try to be more specific in my problem description. My problem that the...
By W.L. Mostia on 9 August, 2018 - 2:17 pm
You need to be more specific in your problem description. Is your problem that the saturation...
Seeking a ground-based sensor system to notify the pilot when a radio-controlled helicopter is hovering precisely over the sensor.
Converting from %Weight to g/L
Existing devices measures concentrations in solution in %Wt units. I have question regarding converting it to g/L units.
By d on 1 August, 2018 - 9:51 am
I started with one liter of pure H2O then added the specified weight of NaOH. Since you are...
By d on 26 July, 2018 - 8:13 am
one liter H2O = 999g / 20 DegC 8% Wt-> 80 grams per liter of water 12% Wt -> 120 grams...
Trying To Find a 4 Wire Pressure Transducer
Posted by Rob Armstrong on 7 July, 2018 - 10:07 pm
Desperately trying to find a replacement pressure transducer!
By Rob Armstrong on 9 July, 2018 - 9:49 pm
THANK YOU Everyone for so much great information!!!!! I have some good photos of the sensor,...
By W.L. Mostia on 8 July, 2018 - 7:32 pm
One of the first things I would do is trace the wiring from the transmitter back to where all the...
By d on 8 July, 2018 - 5:21 pm
Rob, I stand corrected on using a pressure switch with a screw compressor. It would defeat...
By d on 8 July, 2018 - 11:29 am
Get a pressure guage, and add an adjustable pressure switch if you need a pressure cutoff for...
By Rob Armstrong on 8 July, 2018 - 9:07 am
I have a wiring diagram for the existing sensor I can email to anyone if they can help......I...
By Rob Armstrong on 8 July, 2018 - 8:23 am
Thank you for the reply. Unfortunately there are no numbers or labels left on the original...
Fr9E seismic BB5/11 showing erratic reading in Mark VIe till alarm level
on 30 July, 2018 - 2:34 pm
Yaseeen, Here's a link to a good thread, with some good (and bad) comments above and below the...
Natural Gas Fired Boiler-Steam Turbine Basic Q&As
12 year Power generation (wind turbines) power plant operator. Just made Control Operator - hoping to tap into the experience in this group to be better at what I do.
Axial Displacement Probe Calibration
How to get linearity error from actual output voltage/displacement
Replacement Flame Scanner
Where can I get a replacement flame scanner?
Getting a better understanding of gas turbine control
How to get a better understanding of gas turbine control?
Pressure transmitters vs flow meters
By d on 21 July, 2018 - 1:37 pm
Static pressure is a scalar variable, traceable to independent calibration standards. Flow...
Automation Direct DL205 Thermocouple Error
Having issues with thermocouple signals connected to a DL205 PLC. All three are reading several hundred degrees high.
By Steve Myres on 18 July, 2018 - 10:50 am
Might be a data encoding issue. Many 205 modules will send the data in either BCD or binary, and I...
Why SP1 is calculated differently in markVIe?
on 9 July, 2018 - 6:56 pm
khoriwal000, You would need to tell us what the names of the blocks are (such as TTXSPVn, where...
Axial Vibration Monitoring
How to set your counter and normal values for alarm/danger?
Slurry Density Measurement in Ash Wastewater
Posted by M B Ravikumar on 30 June, 2018 - 12:02 am
Looking for a device to measure ash density in wastewater
Smart DP Level Transmitter
Do i need to re-calibrated a SMART level DP transmitter when i change the range in the field?